Evaluation of a closed-loop intravenous total anesthesia delivery system with BIS monitoring compared to an open-loop target-controlled infusion (TCI) system: randomized controlled clinical trial
Intravenous general anesthesia is an anesthetic technique that can be administered with TCI (target-controlled infusion) or closed-loop systems. The authors designed an automatic delivery system using clinical variables such as bispectral index (BIS), heart rate, and blood pressure.
To evaluate the clinical and technical performance of this controller by comparing it to a TCI system.
This was a single-blind, randomized, controlled clinical trial in which 150 patients were recruited: 75 for the TCI group and 75 for the closed loop with BIS. Clinical performance was determined according to the mean percentage of time spent in the BIS range of 40 to 60 during anesthetic maintenance. In addition, adequate intraoperative analgesia, technical performance, intraoperative awakening, and intraoperative recall were evaluated.
The primary outcome showed a mean BIS time between 40 and 60 for the closed loop of 75.24% (± 15.78) versus 59.5% (± 20.3) for the TCI system, with an absolute difference of 15.8%, 95% confidence interval (CI): 9.9 to 21.65, P < 0.0001. The mean time in intraoperative analgesia was 82.4% (25.1) in closed loop and 70.77% (± 32.8) in TCI, with a difference of 4.76 (95% CI: 2.23-21.06), P = 0.016. There was no difference in intraoperative recall.
The closed-loop system was better at maintaining a BIS in the range of 40 to 60 during a general anesthetic than the open system or TCI.
2. Kruger-Thiemer E. Continuous intravenous infusion and multi compartment accumulation. Eur J Pharmacol 1968;43:317-324.
3. Struys MMRF, De Smet T. Evers A, Maze M, Kharasch E. Target controlled infusion and closed-loop administration. Anesthetic pharmacology basic principles and clinical practice 2nd ed. Cambridge, UK: Cambridge University Press; 2011;103-120.
4. Absalom A, Struys MMRF. Absalom A, Struys MMRF. Target controlled infusions. An overview of TCI and TIVA 2nd ed. Ghent, Belgium: Academia Press; 2007;9-40.
5. Hu C, Horstman DJ, Shafer SL. Variability of target-controlled infusion is less than the variability after bolus injection. Anesthesiology 2005;1023:639-645.
6. Mortier E, Struys M, De Smet T, et al. Closed-loop controlled administration of propofol using bispectral analysis. Anaesthesia 1998;538:749-754.
7. Gomez F, Casas F, Fernández J, et al. Total intravenous anesthesia in a closed loop system: Report of the first case in Colombia. Colombian Journal of Anesthesiology 2013;41:306-310.
8. Morley A, Derrick J, Mainland P, et al. Closed loop control of anaesthesia: an assessment of the bispectral index as the target of control. Anaesthesia 2000;5510:953-959.
9. Moerman AT, Herregods LL, De Vos MM, et al. Manual versus target-controlled infusion remifentanil administration in spontaneously breathing patients. Anesth Analg 2009;1083:828-834.
10. Reboso JA, Mendez JA, Reboso HJ, et al. Design and implementation of a closed-loop control system for infusion of propofol guided by bispectral index (BIS). Acta Anaesthesiol Scand 2012;568:1032-1041.
11. De Smet T, Struys MMRF, Neckebroek MM, et al. The accuracy and clinical feasibility of a new Bayesian-based closed-loop control system for propofol administration using the bispectral index as a controlled variable. Anesth Analg 2008;1074:1200-1210.
12. Hemmerling TM, Arbeid E, Wehbe M, et al. Evaluation of a novel closed-loop total intravenous anaesthesia drug delivery system: a randomized controlled trial. Br J Anaesth 2013; 1106:1031-1039.
13. Puri GD, Kumar B, Aveek J. Closed-loop anaesthesia delivery system (CLADS) using bispectral index: a performance assessment study. Anaesth Intensive Care 2007;353:357-362.
14. Hemmerling TM, Charabati S, Zaouter C, et al. A randomized controlled trial demonstrates that a novel closed-loop propofol system performs better hypnosis control than manual administration. Can J Anaesth 2010;578:725-735.
15. Johansen JW, Sebel PS. Development and clinical application of electroencephalographic bispectrum monitoring. Anesthesiology 2000;935:1336-1344.
16. American Society of Anesthesiologists Task Force on Intraoperative Awareness. Practice advisory for intraoperative awareness and brain function monitoring: a report by the American Society of Anesthesiologists Task Force on Intraoperative Awareness. Anesthesiology 2006;1044:847-864.
17. Hemmerling TM, Charabati S, Salhab E, et al. The Analgoscore TM: a novel score to monitor intraoperative nociception and its use for closed-loop application of remifentanil. J Computers 2009;4:311-318.
18. Grinstead CM, Snell JL. Grinstead CM, Snell JL. Central limit theorem. Introduction to probability 2nd ed. Providence, RI: American Mathematical Society Press; 1997;325-355.
19. Varvel JR, Donoho DL, Shafer SL. Measuring the predictive performance of computer-controlled infusion pumps. J Pharmacokinet Biopharm 1992;20:63-94.
20. Schuttler J, Kloos S, Schwilden H, et al. Total intravenous anaesthesia with propofol and alfentanil by computer-assisted infusion. Anaesthesia 1988;43 (suppl):2-7.
21. Glass P, Jacob J, Reves J. Glass P, Jacob J, Reves J. Total controlled infusion. Intravenous drug delivery anesthesia New York: Springer; 1990;367-386.
22. Zhang C, Xu L, Ma Y-Q, et al. Bispectral index monitoring prevent awareness during total intravenous anesthesia: a prospective, randomized, double-blinded, multi-center controlled trial. Chin Med J 2011;12422:3664-3669.